Advanced development setupLink
How to use it?Link
Connect the board
The alphaDelta board should be connected to the kit auxiliary groove connector and the tester board to the alphaDelta. In this way the kit will have acces to both boards writing values in tester board and reading them in the sensor board.
The tester board should be connected between Electrodes A and W from the slot you want to test.
Get the firmware
You will need the latest version of SCK 2.0 Firmware. To enable tester board in firmware uncomment
#define deltaTest in sckAux.h file compile and upload the firmware.
Read the sensorsLink
read Alphadelta [wichSensor: temperature, humidity, 1A, 2A, 2A, 2W, 3A, 3W]
SCK > read alphadelta 1a AlphaDelta 1A: 8.22
Control digital potentiometersLink
control alphadelta 1A set pot [value: 0-100,000]
SCK > control alphadelta 1a set pot 50000 AlphaDelta 1A: set pot 50000 Setting pot to: 50000 Ohms Actual value: 49803 Ohms
Run the testsLink
The tester board is enabled as a control command of Alphasense Delta sensors, so it must be called with the prefix control then the name of the sensor alpha (fuzzy match is supported) then the test keyword followed by the slot number we want to test:
control alpha test 1
To complete the line we should select one of the two test modes:
set followed by a nA value between -1400/+1400. This will instruct the tester board to output that current and verify the readings of the sensor.
control alpha test
SCK > control alpha test 1 set 500 AlphaDelta 1A: test 1 set 500 Setting test current to: 500 Tester Electrode W: 500 Alphadelta 1W: 78.87 Tester Electrode A: 500 Alphadelta 1A: 78.70 Testing finished!
control alpha test 1 full
SCK > control alpha test 1 full AlphaDelta 1A: test 1 full testW,readW,testA,readA -1400,-220.94,-1400,-220.58 -1399,-220.92,-1399,-220.52 -1398,-220.78,-1398,-220.37 -1397,-220.62,-1397,-220.25 -1396,-220.48,-1396,-220.09 ...
How to run the validation?Link
For validating the boards, we propose verifying that: each nA input can only yield a single output in mV in the ADC, at minimum gain (for each electrode).
This means that for a test such as:
control alpha test 1 full
We should obtain an always positively growing value in the readW, readA channels. For this, we have to check that:
If we plot all this results we see that they are normally over 0.1 for all the boards, so we use this value as a minimum threshold, instead of zero:
Finally, this test is also valid for too high gains since once the signal is saturated the \delta value is 0 (saturates in a flat plateau).